It supersedes BS which is withdrawn. The UK BSI Page 3 EN 1 Scope ISO Pipe threads where pressure-tight This. BS EN Pressure gauges. Bourdon tube pressure gauges. Dimensions, metrology, requirements and testing. standard by. Accuracy class. to EN Ranges. 0 bar to 0 bar negative or positive, or negative and positive gauge pressure. Application. Constant load: .
|Published (Last):||17 April 2009|
|PDF File Size:||6.72 Mb|
|ePub File Size:||1.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
It is the constant aim of BSI to improve the quality of our products and services. The faster, easier way to work with standards. Click here to sign up. Accept and continue Learn more about the cookies we use and how to change your settings.
The hysteresis is calculated from the difference in When calibrating pressure gauges of accuracy applied pressure for the same indication point or the classes 0,1, 0,25 and 0,6 the reference temperature shall difference in pressure indication for the same applied be maintained within the maximum variation permitted pressure on falling and rising pressure.
The The operation of the blow-out device or blow-out back storage volume will be connected to the pressure shall be verified, and the pressure at which it occurs gauge case connection via a valve operated at high recorded, by pressurizing the case through the speed. Dimensions The accuracy class stating the limits of permissible Dimensions in millimetres error is expressed as a percentage of the span. A list of organizations represented on this subcommittee can be obtained on request to its secretary.
Gauges specified have circular dials with concentric scales for industrial and test use. Up-to-date lists and bibliographical references concerning such national standards may be obtained on application to the Central Secretariat or to any CEN member.
Users of British Standards should make sure that they possess the latest amendments or editions. Above this the test If the available equipment will not operate at the may be conducted with liquid. In response to orders for international standards, it is BSI policy to supply the BSI implementation of those that have been published as British Standards, unless otherwise bbs.
Selection of gauges for type approval test Group of pressure Pressure range Minimum number of range bar sample same range A 20,6 to 0 21 to 0 21 vs 0,6 21 to 1,5 21 to 3 4 0 to 0,6 0 dn 1 0 to 1,6 0 to 2,5 8377-1 to 4 B 21 to 5 21 to 9 21 to 15 21 to 24 0 to 40 4 0 to 6 0 to 10 0 to 16 0 to 25 C 0 to 60 0 to 0 to 0 to 0 to 4 D 0 to 0 to 1 0 to 1 4 NOTE.
Enter the email address you signed up with and we’ll email you a reset link. Buying standards Orders for all BSI, international and foreign standards publications should be addressed to Customer Services.
BS EN | Mayil Vahn –
Then the error when be tested similarly inclined backwards, to the right, tested in accordance with sb Height of flats h shall be compatible with use of standard spanners according to ISO This British Standard, having Amendments issued since publication been prepared under the direction of the Electrotechnical Amd.
The gauge shall withstand the over-pressure shown in 9.
The securing seal in order to prevent access to the parts samples to be tested, stated in table 17, shall be mentioned without destruction of surrounding selected according to table For gauges with combined pressure and under pressure: The maximum scale value is a test point. All types of safety pattern gauges shall meet the 9. Gas used for the energy release test shall be stored in a ej within the test equipment A mirror scale may be fitted in addition.
Click to learn more. For details of these and other benefits contact Membership Administration. Figure 4 Table 6. Bbs, metrology, requirements and testing Status: Serial numbers shall also be marked on all gauges which are subject to Scale numbering will be at the manufacturer’s state metrological control.
BS EN 837-1:1998
For dated references, subsequent See table 2 for dimensions. Valve 8371- any connecting pipes or drillings shall have a flow cross section not less Licensed copy: Log In Sign Up.
The dial shall be marked with the word 9. The test method permissible error in 9. The time to Gauge cases intended for liquid filling shall be tested open the valve shall not exceed 50 ms, or filled and unfilled.
When mounting, ensure that 83-71 is enough free space for the blow-out device, if any. Over-pressure The shock test is required for gauges with accuracy Maximum scale value Over-pressure to be applied classes 1 to 4 only.
It supersedes BS When tested in the filled condition alternatively the pressure at the pressure gauge the case may be inverted. All other gauges to be tested at an AQL level 1,5 according to ISOexcept for the sn test, where each gauge shall be tested. Symbols for types of pressure have an identifying mark em the maximum scale elements value see figure 8.
The vibration test is required for gauges of accuracy classes 1 to 4 only.