30 Jan Purchase your copy of BS EN , ISO as a PDF download or hard copy directly from the official BSI Shop. All BSI British. ISO is the standard specification and test method for the design of such Luer taper fittings. The two varieties of Luer Taper connections are typically known. ISO All rights reserved. Unless otherwise specified, no part of this . ISO , Conical fittings with a 6 % (Luer) taper for syringes, needles and.
|Published (Last):||18 March 2018|
|PDF File Size:||10.12 Mb|
|ePub File Size:||19.36 Mb|
|Price:||Free* [*Free Regsitration Required]|
ISO 594-1 & 2 Separation Force and Unscrewing Torque Testing for Luer Tapers
Ido is an internal decision for you to make. Discussion on the change. Testing equipment for ISO Leakage; Separation from axial load; Stress cracking; iso 594-1 Non-interconnectable characteristics in the ISO ; Clause 6 specified tests. In essence; this is iso 594-1 just a clerical change of the specification number.
BS EN 20594-1:1994, ISO 594-1:1986
The reference fittings and connectors come in individual boxes and with long-form calibration certificates iso 594-1 an ISO accredited calibration laboratory. Also suitable where ISO Figure 6 is required. We recommend with the 5 Newton weight because: ISO provides test methods iso 594-1 testing the separation force of Luer fitting assemblies and unscrewing torque of Luer fitting assemblies.
Sub-Atmospheric Iso 594-1 Air Leakage 6. This website is best viewed with browser version of up to Microsoft Internet Iso 594-1 8 or Firefox 3. Sometimes the problem gets worse as the replacement specification becomes older.
ISO /1 and ISO /2 REPLACED BY ISO
Learn more about the cookies we use and how to change your iso 594-1. Because of that some product drawings, specifications and procedures reference specification numbers which are obsolete and which have been replaced by new numbers.
For testing of separation force without unscrewing torque, take a look at ISO Visit us at one of our locations. The medical industry, agreeing with Iso 594-1 that it was a good idea, set about designing patient side connectors iso 594-1 are non-interconnectable.
Application of the 5 Newton force is an inaccurate cumbersome operation if the 5 Newton weight is not integral to the Luer Plug Gage. Differences between the two specifications are expected but are technically insignificant to the function of the specified connections. Worldwide Standards We can source any standard from anywhere in the world. The intention is to provide accurate iso 594-1 regardless; errors may exist in the supplied information. The intention is to provide accurate information; regardless; errors may exist in the supplied information.
Here is some sample text for this Global Engineering Directive:.
ISO Fig C.2 Male Reference Luer Slip Connector
Follow this link if you iso 594-1 a more in-depth discussion on the 5 Newton force. Iso 594-1 to learn more. The cost of this service to uso varies depending on how many gages are processed at one time. Live Chat Powered by jivochat.
We also have an open-door policy and welcome you to visit us at one of our locations. The test results can easily be falsified or skewed depending on how much unregulated force is applied to the Iso 594-1 Plug Gauge. ISO Fig C. iso 594-1
For those who just need to know the full details The authors of ISO have made slight changes iso 594-1 the Luer cones which slightly increased their size. Those standards are also being revised, and will eventually refer to the iso 594-1 kso in the ISO series. Give our test engineers a call today for help configuring the best test machine and accessories according to your standard. Rocking shall not be evident 5994-1 the Luer Plug Gage and rigid material fitting undergoing test. Unscrewing torque is tested by applying a 0.
January Replaced By: The price per unit improves with bulk purchases of 3 or more units of the same reference connector. You may experience issues viewing this site in Internet Explorer 9, 10 or Iso 594-1 new ISO specification does not define any gauges. This was essentially a clerical change so iso 594-1 the Needle and Syringe Luer Connectors could be included under the Small-Bore Connector iso 594-1 of specifications: Top Gage Crib Worldwide, Inc.
How do we make our life easier when specification izo change?
There will be an extra cost and the lead-time for the gage may double. This expectation is clearly stated iso 594-1 the ISO Forward. For all tests in ISO requiring an assembled Luer fitting, fittings are assembled using an axial torsion machine, applying a compressive force not greater than If you must have this data certified; this iso 594-1 considered a special request.
Take the smart route to manage medical device compliance.